August 17, 2020
4 MIN READ
When working with high vacuum (HV) and ultra-high vacuum (UHV) systems, there are specific aspects to consider, to make sure they remain efficient and safe.
HV pressure ranges are those between 10 x 10-3 and 10 x 10-7 mbar.
UHV pressure ranges are those between 10 x 10-7 and 10 x 10-12 mbar.
As with all vacuum systems, the established standards, rules, and protocols that govern vacuum factors and matters must be frequently re-examined and re-engineered.
Users should routinely:
Check vacuum levels
Review the pump setup
Perform safety checks
Evaluate measurement methods
Undergo leak detection processes
Related: Successful leak testing and detection underpins the success of any project relying on HV and UHV systems. Download a copy of our Fundamentals of Leak Detection for an easy-to-read introduction to types of leaks, detection methods and more.
HV applications include:
UHV applications include:
Molecular beam epitaxy (MBE)
Related: Find out how aerospace engineers simulate space conditions on Earth in our blog post, How It Works: Vacuum Technology for Space Simulation Chambers. Interested in the HV pumps that make these applications possible? Learn more on our resource page about the Leybold range of High Vacuum Pumps.
Users should carry out a careful assessment of design, materials, and the condition of the vacuum system to identify problem areas. HV and UHV vacuum system efficiency can be improved by following several best practices:
Minimize the chamber’s internal surface area.
Weld only from the inside.
Use low desorption/outgassing rates.
Pre-treat materials — for example, undergo electro-polishing.
Check for internal gaps or trapped volumes, like tapped blind holes.
Use metallic seals.
Reduce the number of seals and feed-throughs.
Outgassing is the process of releasing gas that was dissolved, trapped or absorbed in some material. To create and maintain clean HV and UHV vacuum environments, users must carefully monitor outgassing.
Outgassing typically occurs when materials not usually considered absorbent, release enough molecules to interfere with industrial or scientific vacuum processes. Common sources of outgassing include:
Cleaning surfaces, heating individual components, or performing a bake-out can drive off volatiles.
In HV and UHV conditions, outgassing and degassing levels must be kept as low as possible.
Minimizing effective surfaces reduces the impact of outgassing. The higher the surface area, the greater the outgassing and the higher the system pressure.
A residual gas analyzer (RGA) is a small quadrupole mass spectrometer, typically designed for environment analysis, process control, and contamination monitoring in vacuum systems. RGA's can monitor the quality of the vacuum by detecting (and measuring) minute traces of impurities in a low-pressure gaseous environment.
RGA's are usually mounted directly onto, and into, the vacuum chamber. RGAs perform various functions which would be difficult to obtain by other methods, including: the analysis of various gas-phase reactions, monitoring changes that occur in any gas environment, detecting vacuum leaks, and checking mass flow controllers.
Related: Residual Gas Analyzer or Leak Detector — which one is right for the job? Get the answer in our blog post, When To Use An RGA.
There are many pump types capable of producing HV and UHV vacuum pressure, and most applications require multiple pumps working together. However, pairing different types of vacuum pumps to optimize performance is not always a simple matter.
Several factors impact the choice of pumps, such as:
Noise and vibrations
Initial and ongoing costs
Related: Did you know that high noise levels can leave your team susceptible to hazards? Check out this blog post for 5 Quick Facts About Noise Pollution in Your Lab, and solutions that keep your team safe
Fore pump options include:
Main secondary pump options capable of HV and UHV levels include:
Titanium sublimation pumps
Non-evaporable getter pumps
These pumps produce vacuum conditions by rapidly evacuating, capturing, or tying up gas molecules.
Each pump type has unique advantages and disadvantages, which adds complexity to the selection process.
For example, turbomolecular pumps advantages include:
Easy, kinetic operation
Vibration-generating moving parts
Reduced pumping speed for light gases
Sensitivity to mechanical shock
Buyers should undergo a similar comparison process for each pump type to reach an informed decision.
When creating HV and UHV conditions, it is crucial to consider these three factors: efficiency, working condition considerations, and pump selection. If you're finding selection tough, check out our blog post, How Do I Know Which Vacuum Pump Is Right For Me? With the help of resources like these, an approach that focuses on the key elements of working condition considerations, efficiency and pump selection will keep you on the path to delivering optimum performance and reliable HV and UHV conditions.
And if you've got questions, reach out to the Leybold team! Click the button below and start the conversation. We're always happy to hear from you and can help with all your HV and UHV needs.
Download our e-Book "Fundamentals of Leak Detection" to discover leak detection essentials and techniques.